1. The 50KHz-300MHz frequency range of si5351 direct output provides better than 70dB dynamics, the extended 300M-900MHz frequency band provides better than 60dB dynamics, and the 900M-1.5GHz frequency band provides better than 40dB dynamics.
2. The unit design is suitable for the original equipment, the form and function are simple and practical PC control software, you can export Touchstone (snp) files for various radio design and simulation software through the PC software.
3. Leverage Spectra Premium's OE engine management system design expertise to meet requirements. Platinum connector threads reduce the number of connection points, allowing faster response to changes. The printed circuit board made of ceramic is protected by silicone and has excellent resistance to temperature changes and vehicle vibrations.
4. It includes a 2.8-inch touchscreen and a 500mAh battery for 2 hours of offline use. For longer measurements, please connect an external power source or contact me to make a larger battery version. The TX/RX method can measure the complete S11 and S21 parameters. If you need to obtain S12 and S22, you need to manually replace the transceiver port wiring.
Name: Vector Network Analyzer Kit
PCB: 54mm x 85.5mm x 11mm (without connectors and switches)
Measurement frequency: 50KHz ~ 300MHz (50KHz -900MHz, enable extended firmware)
RF output: -13dbm (-9dbm)
Frequency accuracy: <0.5ppm
Measurement range: 70dB(50kHz-300MHz),(50dB(300M-600MHz), 40dB(600M-900MHz) with extended firmware enabled);
Port SWR: < 1.1
Display: 2.8 inch TFT (320 x240)
USB interface: USB type-C Communication mode: CDC (serial)
Power supply: USB 5V 120mA, built-in 400mAh battery, charging current 0.8A
Number of calibration points: 101 (fixed)
Number of scan points: 101 (fixed)
1 x NanoVNA Main Unit
1 x USB Type-C Data Cable
2 x 30mm SMA Male to Male RG174 RF Cables
2 x SMA Male Calibration Parts
1 x SMA Female to Female Connector
The color deviation is subject to actual system configuration and 1-3cm slight dimension difference is allowed due to manual measurement.